Comparison of Fit for Sealed and Loose-Fitting Surgical Masks and N95 Filtering Facepiece Respirators.
subject: surgical mask, Respiratory protection, fit test, breathability, N95 filtering facepiece respirator, inward leakage
description: N95 filtering facepiece respirators (N95 FFRs) and surgical masks are comprised of multiple layers of nonwoven polypropylene. Tight-fitting N95 FFRs are respiratory protective devices (RPDs) designed to efficiently filter aerosols. During the COVID-19 pandemic, health care workers (HCWs) throughout the world continue to face shortages of disposable N95 FFRs. Existing version of widely available FDA cleared loose-fitting surgical masks with straps do not provide reliable protection against aerosols. We tested the faceseal of a modified strapless form-fitting sealed version of surgical mask using quantitative fit testing (QNFT) and compared the performance of this mask with that of N95 FFRs and unmodified loose-fitting surgical masks. Twenty HCWs participated in the study (10 women; 10 men; age 23-59 years). To create the sealed surgical masks, we removed the straps from loose-fitting surgical masks, made new folds, and used adhesive medical tape to secure the new design. All participants underwent QNFT with a loose-fitting surgical mask, the sealed surgical mask, and an N95 FFR; fit factors were recorded. Each QNFT was performed using a protocol of four exercises: (i) bending over, (ii) talking, (iii) moving head side to side, and (iv) moving head up and down. When the overall fit factor for the sealed surgical mask or N95 FFR was <100, the participant retook the test. Participants scored the breathability and comfort of the sealed surgical mask and N95 FFR on a visual analog scale (VAS) ranging from 0 (unfavorable) to 10 (favorable). The median fit factor for the sealed surgical mask (53.8) was significantly higher than that of the loose-fitting surgical mask (3.0) but lower than that of the N95 FFR (177.0) (P < 0.001), equating to significantly lower inward leakage of ambient aerosols (measuring 0.04-0.06 µm) with the sealed surgical mask (geometric mean 1.79%; geometric standard deviation 1.45%; range 0.97-4.03%) than with the loose-fitting surgical mask (29.5%; 2…
creator: Karunakaravel Karuppasamy, Nancy Obuchowski
source: Annals of work exposures and health
date: 2021